How A Mass Spectrometer Works & How Our Leases Save You Time & Money
Whatever your mass spectrometer requirements and needs might be, our lease program can meet them. We lease a wide variety of models and brands. Reach out to learn more and receive a lease estimate today.
All equipment brands/models are available
Excedr’s Mass Spectrometer Leasing Program Benefits:
- Eliminates the upfront cost of purchasing equipment by spreading its cost over time
- Minimizes equipment downtime with included complete repair coverage and preventive maintenance
- Takes advantage of potentially 100% tax deductible* payments, providing you significant cash-savings
- Expedites the administrative work needed for instrument procurement and logistics
- Conserves working capital, enabling you to reinvest in your core business and operations (staffing, inventory, marketing/sales, etc.)
- Accommodates all manufacturer and model preferences
*Please consult your tax advisor to determine the full tax implications of leasing equipment.
Scientists and researchers use mass spectrometry (MS) as an analytical tool to determine the relative abundance of isotopes, identify unknown compounds, quantify known components, and elucidate structures found in both pure samples and complex mixtures.
Not to be confused with mass spectroscopy—the term’s use is discouraged due to the possibility of confusion with light spectroscopy—MS measures the distribution of ions by mass, or mass-to-charge ratio (m/z), of molecules (and/or its fragments) and atoms by ionizing the atoms and molecules with a high-energy electron beam and then deflecting the ions through a magnetic field based on their mass-to-charge ratios (m/z). These measurements can often be used to calculate the molecular weight of the sample.
A mass spectrometer—the device that utilizes MS—consists of an ion source, a mass analyzer, and a detector. Typically, as a sample is introduced into the spectrometer, it is bombarded with electrons and ionized, allowing the compounds to become charged and fragmented.
Using the mass analyzer, the ions are separated by an electromagnetic field according to their mass-to-charge ratio (m/z). The detector system then ascertains the relative abundance of each of the resolved ionic species.
The spectrometer will then create a graphical representation of the ion’s specific ion abundance vs. m/z that were separated during analysis. This plotting is referred to as the mass spectrum. It can represent many different types of information based on the type of mass spectrometer and the specific experiment applied. These spectra are used to determine the elemental or isotopic signature of a sample, the masses of both particles and molecules, and to elucidate the chemical structure or identity of molecules and chemical compounds.
MS is used in a diverse range of fields that includes proteomics, metabolomics, forensic toxicology, as well as general clinical research. It has become a particularly powerful tool used in proteomics as well. Using the technique enables researchers to perform proteome-wide analysis and characterization of proteins from a wide range of cell types and organisms.
Components & Benefits
Mass spectrometer instrumentation is made up of three major components that each carry out the fundamental steps in the analytical process: ion generation, ion separation, and ion detection.
Ionization is used to produce gas phase ions suitable for resolution in the mass analyzer. In order for a mass spectrometer to ionize a sample, or analyte, under analysis, an ion source is required. This ion source is how a mass spectrometer system generates ions. Afterward, the ions are transported via magnetic or electric fields directly to the mass analyzer.
Ionization techniques have played a vital role in determining what types of samples can be analyzed using MS. Depending on what you are analyzing, you’ll have to use specific types of ionization. Electric and chemical ionization are used for both vapors and gas, while matrix-assisted laser desorption/ionization (MALDI) and electrospray ionization are used for solid or liquid biological samples.
There are two classifications of ionization used in MS.
Hard & Soft Ionization
The choice of ionization source is extremely important, and is usually classified as “hard” or “soft”.
Hard ionization fragments molecules, whereas soft offers little or no fragmentation. One example of hard ionization is electric ionization, or EI. This technique creates a high degree of fragmentation and yields in-depth mass spectra that can provide insightful, critical information. This type of ionizing tend to have lower m/z value compared to the molecular mass.
On the other hand, soft ionization includes processes such as chemical ionization (CI), electrospray ionization (ESI), and MALDI.
The mass analyzer is used for separating or resolving the ions according to their mass-to-charge ratio (m/z). This component is how the MS system separates ions.
There are six general types of analyzers used in MS:
- Time-of-flight (ToF)
- Ion trap
- Magnetic sector
- Electrostatic sector
- Ion cyclotron resonance
Each type has its advantages and disadvantages. While some offer high resolution and dynamic range, they can be expensive and bulky. Others may offer high sensitivity and reproducibility but lack the scan speed you require.
No single analyzer is suited for all analyses, and the advantages or limitations can have a significant impact on your lab. Therefore, understanding the different principles, features, and characteristics of each type is essential to choosing the one that best suits your needs and provides you optimal capabilities.
A key element of any system, the spectrometer’s detector is used to detect separated ions and convert them into a measurable signal. The separated ions are measured, and those measurements are sent to a data systems that creates a mass spectrum, a graphical representation of each ion’s m/z ratios and their relative abundance, plotted against their intensities.
There are several types of detectors available. The type you use depends on a variety of factors that can include spatial information retention and dynamic range, as well as spectrometer design and application requirements. They include:
- Photoplate detectors
- Faraday detectors (or Faraday cups)
- Electron multipliers (EM) or secondary electron multipliers (SEM)
- Daly detectors
- Image-current detection technology
- Focal-plane array detectors
Many of the various types of detectors offer high sensitivity, as well as linear and quantitative response. However, some detectors are designed for very specific functions. Noise, uniformity, and counting single or multiple masses are also important considerations when choosing an instrument.
Ion detection technology has not received as much attention as the ion source and mass analyzer. Traditional analog (Faraday detectors)—and electron multiplier (EM) detectors have been used for decades, and although advances in ion detection technology have occurred, they have been incremental. Nonetheless, it is an area of importance, due to the fundamental need to detect ions.
A sample can be introduced into an MS system in several ways, depending on the experimental setup.
For many researchers and scientists in the life sciences, MS almost always means liquid-chromatography-mass spectrometry (LC-MS), gas chromatography-mass spectrometry (GC-MS), inductively coupled plasma-mass spectrometry (ICP-MS), as well as several other integrated analytical solutions.
Because mass spectrometers typically lack any analytical front end, integrating these systems with chromatographic technologies has become fundamental to analytical research. However, stand-alone devices can still be found in use for the analysis of pure solid and liquid compounds, or as chemical-specific analyzers that do not require separation from each other or the matrix they occur in.
Furthermore, tandem mass spectrometry, or MS/MS, is used to increase the overall ability of the two devices to analyze chemical samples. These coupled systems can be used alone as well, or together with GC or LC systems.
Mass spectrometry is used in a variety of applications across a wide range of fields. It’s applications commonly include:
- Proteomics: protein identification and characterization, peptide sequencing, and mapping numerous organelles.
- Metabolomics: metabolic fingerprinting and profiling, lipodomics studies, cancer screening and diagnosis, and biofuels generation and use.
- Pharmaceutical Analysis: drug testing and discovery, drug metabolism, drug degradation, product analysis, preclinical development, and pharmacokinetic and pharmacodynamic analyses.
- Clinical Diagnostics: therapeutic drug monitoring, biomarker discovery and profiling, disease detection and risk assessment, and direct analysis of biological samples.
- Forensic analysis: trace, fiber, and hair analysis, arson investigation, explosives and gun shot residue, and drug identification and abuse confirmation.
- Environmental Analysis: pesticide screening and quantitation, tap water testing, soil contamination analysis, and carbon dioxide and pollution monitoring.
The Introduction of the Triple Quadrupole Mass Spectrometer
Richard A. Yost and Christie G. Enke’s report on the triple quadrupole mass spectrometer in the Journal of the American Chemical Society in 1978 has been cited as the birth of mass spectrometer instruments.
However, the pair did not set out to create a mass spectrometer when they began to attempt entirely computerizing an analytical instrument. The triple quadrupole mass spectrometer, over 40 years old now, enabled scientists to use MS as an analytical chemistry technique. Before Enke and Yost’s work, MS was primarily used by organic and physical chemists. Since the pair’s report, the triple quadrupole mass spectrometer has turned into the workhorse of quantitative mass spectrometry.
Reduce Your Upfront Costs by Leasing an LC-MS System
Choosing from a wide range of techniques and technologies available can be a daunting task, and it is essential to keep scientific application, performance expectations, and budget in mind. Doing so will help you procure the exact mass spectrometer system to match you needs. Regardless of whether you’re interested in an ICP-MS, LC-MS, or a standalone MS/MS, we can help.
This off-balance sheet financing structure provides three options at the end of the term. The lessee has the option to return the equipment to the lessor, renew at a discounted rate, or purchase the instrument for the fair market value. Monthly payments are also 100% tax deductible which yields additional monetary savings.
If you recently bought equipment, Excedr can offer you cash for your device and convert your purchase into a long-term rental. This is called a sale-leaseback. If you’ve paid for equipment within the last ninety days, we can help you recoup your investment and allow you to make low monthly payments. This also frees up money in your budget rather than tying it down to a fixed asset.
Mass Spectrometer Manufacturers & Models on the Market
Ultivo Triple Quadrupole LC/MS, 6495C Triple Quadrupole LC/MS, 6470B Triple Quadrupole LC/MS, 6545XT AdvanceBio LC Q-TOF, 6530 Q-TOF LC/MS, 6545 LC/Q-TOF, 6550 iFunnel Q-TOF , 6560 Ion Mobility LC/Q-TOF, StreamSelect, RapidFire 400, LC/MSD, LC/MSD XT, LC/MSD iQ, 6230B Time Of Flight LC/MS
ACQUITY QDa, SQ Detector 2, Xevo TQ-XS, Xevo TQ-S, Xevo TQ-S Micro, Xevo TQ-S Cronos, Xevo TQD, XEVO TQ-GC, SYNAPT, MALDI SYNAPT
TripleTOF, TripleTOF 6600+ System, TripleTOF 6600 System, TripleTOF5600 System, X500B QTOF, X500R QTOF, QTRAP LC-MS/MS Systems, QTRAP 6500+ System, Triple Quad 5500+ LC-MS/MS System, QTRAP 4500 System
AxION 2 TOF, AxION DSA, Torion T-9 Portable GC/MS, QSight SP50 Online SPE System, QSight Pesticide Analyzer, QSight Triple Quad LC/MS/MS, Clarus SQ 8 GC/MS
GCMS-TQ8050 NX, GCMS-TQ8040 NX, GCMS-QP2020 NX, GCMS-QP2010 SE, LCMS 9030 Q-TOF, LCMS-8060NX, LCMS-8050, LCMS-8045, LCMS-8040, LCMS-8030, LCMS-2020, MALDI-7090, AXIMA MALDI-TOF Systems, AXIMA Assurance
rapifleX, rapifleX MALDI TissueTyper, rapifleX MALDI PharmaPulse, microfleX Series, autofleX maX, ultrafleXtremeMALDI Biotyper Systems, timsTOF, maXis II, maXis II ETD, amaZon speed, amaZon SL, EVOQ
TSQ Altis, TSQ Quantis, TSQ Quantiva, TSQ Endura, TSQ Fortis, TSQ Quantum, Orbitrap Fusion, Orbitrap Exploris, Q Exactive, Exactive Series, LCQ Fleet, LTQ XL Linear, Velos Pro, ISQ EC, MSQ Plus, ISQ LT, ISQ QD, Delta V IRMS, 253 Plus, Argus VI SVMS, Helix MC Plus, Helix SFT, Neptune Series, Triton Series TIMS, ELEMENT GD PLUS, ELEMENT Series HR-ICP-MS,iCAP RQ ICP-MS, iCAP TQ ICP-MS, iCAP TQs ICP-MS